Emission power up to 210W. Shorter exposure time with high power. Achieve stable, accurate and quick photography of different tooth positions.
Various Photographic Methods - Easy to operate. Built-in remote control receiving circuit and holder interface, can be connected to the wired remote controller and triangle holder. A varity of photographing methods available, ensuring safer photographing and easier operation. 0.4mm focus X-ray tube.
Provide high-quality and clearer image for various diagnostic needs. 15000mAh high-capacity power lithium battery delivers a longer endurance. More than 1,500 dental films can be taken under a full charge. Constant DC high frequency and voltage current output. Ensure that X-ray output is not affected by battery voltage fluctuations.
Unique internal design and lead shield. Efficiently minimize the X-ray leakage.
Maximize the protection of operators from ionizing radiation. 2.8 Inch capacitive touch screen.Woodpecker Ai Ray Portable X-Ray Machine. Is a compact & lightweight device, that ensures the safety of both dentist & patient by minimal radiation. It produces a clear, higher-quality image with a better shooting effect through an inbuilt Canon tube and a shorter exposure time. Ai Ray Portable X-Ray Offers huge rechargeable battery enables easy & multiple shooting keeping everywhere. This X-ray unit is compact in size and its portable nature helps you to use it in a comfortable manner.
Moreover, this unit is highly ensured with safety standards owing to its minimum exposure time. High efficiency for sharp radiography gives high-quality images. Non-invasive diagnostic imaging equipment taking care of the patient's safety to radiation exposure. All X-rays are compatible with IOPA Film and RVG leading to the convenience of operations.
Maximum power: 0.21kw(70kV, 3mA, 0.1s). Nominal electric power: 0.21kw(70kV, 3mA, 0.1s).
Tube voltage output is fixed at 70kV, error ±10%. Tube current output is fixed at 3mA, error ±20%. The exposure loading time adjustment range is 0.02s~2s, the grade is adjustable, the grade is selected according to R'10 numerical system; with deviation ±(10%+1ms). Distance from focus to skin. 1 x Ai Ray Portable X-Ray Machine.1 x Power adapter & Cable. Take out all the parts from the packing box. Be careful not to drop or damage the equipment. Take out the power adapter and power line from the packing box and connect them. Only the power adapter and power line provided with the equipment can be used. The user of the medical device must comply with the requirements of the relevant operating regulations and relevant regulations of the medical department, and is limited to the use of trained doctors or technicians. Turn on the Dental X-ray power switch, the LCD screen lights up, accompanied by a beeper "di" sound prompt 2. Check the battery of the equipment to ensure the normal operation of the equipment 3. Select the human body, tooth position and the equipment mode 4. The system has a default exposure time, or adjust the shooting time as required 5. Prepare film or image plate scanner (IP image plate) or Digital Intraoral X-ray Imaging System (sensor). A high-quality equipment of image receptor (film or IP image plate or sensor) in a sealed protective bag will be put in the patient's mouth, parallel to the longitudinal axis of the tooth. The effective surface of the equipment of image receptor is facing the tooth; 2.
Move the Dental X-Ray Device to the teeth on the patient's face and adjust the position of the equipment and the patient according to the angle displayed on the screen; 3. Ensure that the light cone of the equipment of image receptor is perpendicular to the position of the IP image plate, press the X-ray machine shooting switch; the exposure key should be pressed for the whole process until the "di" sound prompt of the beeper occurs; 4.
When the exposure is finished and the image is taken successfully, remove the equipment of image receptor from the patient's mouth. When different modes of the equipment, tooth positions and human bodies are selected, the control panel automatically displays the exposure time. Select the required equipment of image receptor, and the corresponding area of the equipment is displayed in blue. After the selection is successful, it will automatically exit to the interface, and the device selection icon will change to the corresponding device icon. After selecting the equipment mode, select human body mode.
Different human body models can be selected according to the age of the patient. After the selection is successful, the human body model area will display the corresponding options. Select the tooth type to be shot, and the corresponding area of the tooth is displayed in blue.After the selection is successful, it will automatically exit to the interface, and the tooth position selection icon will change to the corresponding tooth position icon. Different languages and factory modes can be set. After the setting is successful, it will automatically exit.
The exposure time can be adjusted from 0.04 seconds to 1 second in 15 steps. Long press the exposure switch of Dental X-Ray device to carry out exposure. Long press the exposure button on the exposure handbrake to carry out exposure. During exposure, the exposure indication light of the exposure handbrake will be on. Connect one end of the charger to the charging port of the equipment and the other end to the power supply of the network (100-240V, 50/60Hz); 2.When charging, the equipment displays the charging icon, and when charging is finished, the battery is charged fully; 3. Disconnect the power supply and the charger when the charging is complete; 4. A single charge takes about 4 hour. This item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies.